It also admits other uses according to doctor’s judgment due to its length and designing.
The Stening® “Y” has been used with success in tracheotomised patients and in mechanical respiratory assistance, in combination with tracheostomy cannula, to allow ventilation in seriously ill patients when other methods are not possible.
The process will take place under general anaesthesia. This kind of implants must be carried out by experienced staff. The stent can be fitted on a special forceps that is used for “Y” prostheses implant.
Lubricate the forceps’ margin with lidocaine gel. Introduce it inside de stent in order to make its valves penetrate into its bronchial branches (see the photo on previous page). Ventilate the patient with oxygen until you reach the higher saturation possible. After that we will proceed to the patient extubation by removing the tracheoscope from the airway. Immediately, and with the laryngoscope help, the forceps will be guided towards the trachea. When you close the forceps’ valves the bronchial branches from the stent will join, and, in this position, it will pass through the vocal chords towards the trachea. The manoeuvre will continue to displace the forceps-stent assembly inside the trachea until it approaches to the carina. When the edge of the forceps-stent assembly is close to the tracheal carina, the valves must open kindly so that they can notice the prosthesis arrival to the tracheal bifurcation.
At this moment, the valves will open completely so that the stent bronchial branches can be introduced inside the source bronchus. Then the button from the forceps that holds the stent against the carinal edge will be pushed at the same time the forceps is removed.
The manoeuvre must be fast, knowing that it is carried out while the patient is extubated and in apnoea.
To make sure that the stent does not go to the oesophagus and that it cross the vocal chords safely, use direct optic vision.
To verify if the stent passes through the glottis, the optic must accompany the forceps in a parallel way during the manoeuvre.
This is only possible if a second surgeon holds the laryngoscope in a suitable intubation position, whilst the bronchoscopist holds the forceps with his right hand and the optic with his left hand.
Other ways of implant are possible; for instance, the insertion of a bended prosthesis inside a bronchoscope with the right size, then pushing the stent through the bronchoscope’s interior with a alligator forceps or a smaller diameter bronchoscope when it is closer to the tracheal carina.
The manoeuvre can be completed by adjusting the prosthesis with a forceps.
We will proceed to the patient’s intubation with a tracheoscope.
The extraction is easier.
The stent must be taken by its proximal border with a strong forceps to remove it kindly by pulling the forceps and removing the prosthesis at the same time you remove the tracheoscope.
The product should not be reused because this can cause cross contamination.